J. Synchrotron Rad. (2013). 20, 683-690 [ doi:10.1107/S0909049513021201 ]
Abstract: The conventional energy calibration for nuclear resonant vibrational spectroscopy (NRVS) is usually long. Meanwhile, taking NRVS samples out of the cryostat increases the chance of sample damage, which makes it impossible to carry out an energy calibration during one NRVS measurement. In this study, by manipulating the 14.4 keV beam through the main measurement chamber without moving out the NRVS sample, two alternative calibration procedures have been proposed and established: (i) an in situ calibration procedure, which measures the main NRVS sample at stage A and the calibration sample at stage B simultaneously, and calibrates the energies for observing extremely small spectral shifts; for example, the 0.3 meV energy shift between the 100%-57Fe-enriched [Fe4S4Cl4]= and 10%-57Fe and 90%-54Fe labeled [Fe4S4Cl4]= has been well resolved; (ii) a quick-switching energy calibration procedure, which reduces each calibration time from 3-4 h to about 30 min. Although the quick-switching calibration is not in situ, it is suitable for normal NRVS measurements.
Keywords: nuclear resonant vibrational spectroscopy; high-resolution monochromator; in situ energy calibration; quick-switching energy calibration; small energy shift; energy scale; energy position.
Portable Document Format (PDF) file
To open or display or play some files, you may need to set your browser up to use the appropriate software. See the full list of file types for an explanation of the different file types and their related mime types and, where available links to sites from where the appropriate software may be obtained.
The download button will force most browsers to prompt for a file name to store the data on your hard disk.
Where possible, images are represented by thumbnails.
Copyright © International Union of Crystallography