research papers
Xray collimation by crystals with precise parabolic holes based on diffractive–refractive optics
^{a}Institute of Physics, Academy of Sciences of the Czech Republic, v.v.i., Na Slovance 2, CZ18221 Praha 8, Czech Republic, ^{b}Swiss Light Source, Paul Scherrer Institut, CH5235 Villigen, Switzerland, and ^{c}Department of Condensed Matter Physics, Faculty of Science, Masaryk University, Kotlářská 2, CZ61137 Brno, Czech Republic
^{*}Correspondence email: peter.oberta@psi.ch
Two crystals with precise parabolic holes were used to demonstrate sagittal beam collimation by means of a diffractive–refractive doublecrystal monochromator. A new approach is introduced and beam collimation is demonstrated. Two Si(333) crystals with an asymmetry angle of α = 15° were prepared and arranged in a dispersive position (+,−,−,+). Based on theoretical calculations, this doublecrystal setup should provide tunable beam collimation within an energy range of 6.3–18.8 keV (Θ_{B} = 71–18.4°). An experiment study was performed on BM05 at ESRF. Using 8.97 keV energy, the beam profile at various distances was measured. The experimental results are in good agreement with theoretical predictions. Owing to insufficient harmonic suppression, the collimated (333) beam was overlapped by horizontally diverging (444) and (555) beams.
Keywords: collimation; diffractive–refractive optics; Xray optics; parabolic hole.
1. Introduction
Beam shaping based on diffractive–refractive optics has been previously studied and demonstrated by our group at the Institute of Physics of the Academy of Sciences of the Czech Republic. Both Bragg (Hrdý, 1998) and Laue (Hrdý et al., 2006) diffraction approaches have been demonstrated. Based on theoretical calculations (Hrdý & Oberta, 2008) and raytracing simulations (Artemiev et al., 2004), various applications for diffractive–refractive optics were proposed and tested at synchrotron facilities. Recently, the smallest focal spot size by diffractive–refractive optics (Oberta et al., 2010) was achieved and a novel method of higher harmonics separation in space was proposed at the Swiss Light Source (SLS; Hrdý et al., 2011). In addition to beam focusing, another application of diffractive–refractive optics is beam collimation. In this paper a new approach is introduced and beam collimation is demonstrated using a dispersive arrangement of two crystals with precise parabolic holes.
There are a number of methods used to collimate Xrays. Mirrors or asymmetrically cut crystals can be used (Mori & Sasaki, 1995; Renninger, 1961), but these approaches change the direction of beam propagation and additional optics are needed to correct this deviation. Our method is based on two asymmetrically cut crystals into which we machined holes with a precise parabolic profile: two channelcut crystals. The dispersive arrangement has the advantage, compared with other methods, that it conserves beam direction: the entrance and exit beam positions are fixed. The same advantage holds for refractive lenses (Snigirev et al., 1996); however, they can only be used for hard Xray radiation. Another advantage of the diffractive–refractive method is lower loss compared with other collimation methods. The Si(333) reflection loses just about 10% of after diffraction.
2. Theoretical description
Imagine a crystal into which a hole is drilled. This hole is further machined to reach an ideal parabolic shape. The crystal is cut in such a way that the crystallographic planes of the (111) orientation form an angle α with the crystal surface (asymmetric crystal), Fig. 1. If one takes a point source, located at a distance S from the crystal, then the diffracted radiation from the parabolic surface profile of the drilled hole in the crystal will be sagittally focused to a point at a distance f from the crystal. The formula describing the parabolic profile can be expressed as
where a is the parabola parameter. Then the relationship between S, f and a can be expressed as (Hrdý & Oberta, 2008; Snigirev et al., 1996)
where
where for silicon
Here r_{e} is the classical electron radius, F_{0} is the V is the unitcell volume, λ is the wavelength, d_{hkl} is the atomic plane spacing, α is the asymmetry angle (α > 0 for the grazingincidence case), θ is the N is the number of diffraction events and b is the asymmetry factor. A plot of formula (2) for d_{hkl} = (111) shows the dependence of the focusing distance over energy or (Oberta et al., 2010). By using only one crystal with a parabolic shaped hole, Fig. 1, the number of diffracting events is two, N = 2. As shown by Hrdý & Siddons (1999), such a crystal arrangement cannot achieve a sharp focal spot, because of the combined vertical and horizontal spread of the beam. To cancel the vertical and horizontal beam spreads, a Bartels crystal arrangement must be used. For this case the number of diffracting events is four, N = 4 (Fig. 2). A single crystal with two diffracting events, N = 2, is a nondispersive system, but the twocrystal system used in the experiment represents a dispersive system, with N = 4 diffracting events (Fig. 2).
The focusing condition is sensitive to dispersion and thus depends on the reflection. Fig. 3 shows the focusing distance over a of a dispersive arrangement of two crystals with a parabolicshaped drilled hole for the Si(333) crystal orientation. The crystal–source distance was S = 33613 mm as used at BM05 at ESRF where the experiment was performed. The parabola parameter of our crystals is a = 0.15 mm^{−1}. As can be seen from Fig. 3, for certain Bragg angles (energies) the focusing distance is several hundred meters from the crystal or becomes infinite (inclined dashed area). Under these conditions the crystal arrangement is a collimator. The collimation is therefore independent of the divergence of the impinging beam. This is the working principle of the diffractive–refractive Xray collimator.
Fig. 3 plots the calculated focal distance for three different harmonics. The black full line represents the Si(333) focusing versus dependence. We can see higher harmonics, plotted as red dashdotted [Si(444)] and blue dashed [Si(555)] lines. A negative focusing distance means a divergence of the beam with a focal point placed before the crystals. A positive focusing distance represents focusing with the focal point placed after the crystals. If a focusing distance of an optical setup, like the proposed crystal arrangement, is from several hundred meters up to kilometers; such a setup can be described as collimating. For the case of the diffractive–refractive Xray collimator this value was set to ±500 m. The focusing distance is very sensitive to the crystal–source distance S (Fig. 4). By changing S and moving the crystals over a range of 10 m, we can achieve a collimation effect over an angular range of Θ_{B} = 71–18.4°, that corresponds to an energy range of 6.3–18.8 keV. Because the proposed crystal collimator is based on the diffractive–refractive effect, which is much stronger for longer wavelengths, the working range of the collimator lies in the soft Xray region. In Fig. 4 we find two collimating angular ranges (inclined dashed area). The range for larger Θ_{B} (smaller energies) is broader than that for smaller Θ_{B} (larger energies). Both ranges are approaching each other with increasing distance S.
Table 1 includes the tabulated angular and energy ranges for eight different crystal–source distances, along with the focal distances of some harmonics. Table 2 also includes the tabulated broadening of the beam compared with an ideal collimation as a percentage of the impinging beam dimensions. The additional broadening of the Si(333) diffraction for a focal distance of 500 m is within ∼0.18% of ideal, which justifies our approximation approach. In the case of the higher harmonics the share to the beam broadening is higher, but still in the range between 5.8% and 14%.


As one can see from Fig. 4 and Table 2, harmonic contamination will project itself as an additional beam broadening and will decrease the multiwavelength collimation. One can overcome this problem by using a special crystal geometry to separate the higher harmonics in space without deforming the reflectivity curve, as carried out by traditional harmonics separation methods (Hrdý et al., 2011), or by introducing a slit system between the two crystals.
3. Experimental results
The experiment was performed on a vertical diffractometer at the BM05 beamline at ESRF. We set two crystals with precise machined parabolic holes into a dispersive position and set the beamline monochromator to 8.97 keV.
First we set only one crystal into the diffraction position (Θ_{BSi(333)} = 41.4°) and detected the diffraction. Then we removed the crystal and set the second crystal. After adjusting the second crystal we set both crystals inline. At different distances we recorded diffraction images on Xray films and on a digital camera.
3.1. Singlecrystal arrangement
In a setup with only one crystal diffracting we detected higher harmonics. As can be seen in Fig. 5, there are two parabolicshaped diffraction spots on the Xray film. The full parabolic diffraction spot corresponds to the Si(333) diffraction, which copies the shape of the parabolic groove. Based on the distance of 7 m of the Xray film and the angular separation of 0.7° of the two diffractions one can calculate the corresponding energy of the other diffraction. Based on this approach we found out that the partial parabolic diffraction corresponds to the Si(444) diffraction energy.
After diffraction from one crystal the higher harmonics are separated in space (Hrdý et al., 2011), and after two crystals the higher harmonics are assembled again inline and overlap each other. Therefore, higher harmonics contamination could be solved by introducing a slit system between the two crystals. The schematic in Fig. 5 shows the layout of the experiment. Here A is the crystal–source distance (33600 mm), B is the crystal–Xray film distance (7000 mm), X is the Si(333) diffraction spot size [width of the Si(333) parabolic diffraction, 9.05 mm] and Y is the width of the primary beam, 10.9 mm. These dimensions are related as
From this simple relationship we can calculate the width of the Si(333) collimated diffraction spot if the collimation is ideal. For the ideal collimation the Si(333) diffraction spot is 9.02 mm. The error between experiment (9.05 mm) and calculation (9.02 mm) is just 0.3%.
3.2. Doublecrystal arrangement
By assembling the two crystals inline, higher harmonics, which were spatially separated by the first crystal, are assembled again together by the second crystal. This way the spot after the two crystals is contaminated by higher harmonics, which are not collimated and degrade the apparent collimation effect. Also the parabolicshaped diffraction after one crystal is canceled by the second crystal. As a result we restore a line beam (Figs. 6 and 7). By rocking one of the crystals (Fig. 6), higher harmonics are visible. The diffraction in Fig. 6 has a FWHM of 10.4 arcsec, which corresponds to the Si(444) diffraction at 8.97 keV with a theoretical FWHM of 10.6 arcsec. The second diffraction spot, visible in frames 3–8, is the diffracted Si(333). Fig. 7 shows an Xray film image of diffraction after two crystals at a distance of 7 m after the crystals. The false colors represent the different energies. The red color denotes the diffraction spot of the Si(333) diffraction, green the Si(444) diffraction, and light blue for the Si(555) diffraction. The dimensions of the diffraction footprint in Fig. 7 correspond to the theoretical spread of the higher harmonics calculated from Fig. 3. The theoretical broadening of the beam after a slit size of 5 mm should be 5.16, 5. 48 and 5.67 mm for the Si(333), Si(444) and Si(555) diffraction, respectively. The beam dimensions obtained by experiment were 5.12, 5.44 and 5.70 mm for the Si(333), Si(444) and Si(555) diffraction, respectively. The theoretical description (Hrdý & Oberta, 2008; Snigirev et al., 1996) supposes that all the diffraction events occur in one point in space; this way a clear crystal–source distance is set. In reality there are four different points of diffraction in the two crystals; the diffracted beam bounces twice from each crystal. Contrary to theory, we do not have one spatial point for all four diffracting events and one crystal–source distance, but four points and four different crystal–source distances. This situation can be simplified to one of the three different points in Fig. 2. We can either choose the first diffraction event at the first crystal (A) as the point where all diffraction events occur, or we choose the very exit point of the fourth and the final diffraction on the second crystal (C). We can also choose a point between the two crystals (B). At the end we can calculate using three different points, which gives us three different crystal–source distances (Fig.2). This situation leads to aberrations and thus disagreements between calculation and experiment (Hrdý et al., 2005).
The jagged shape of the lefthand side of the diffraction spot in Fig. 6 is due to crystal subsurface imperfections. The same imperfection can be seen in Fig. 7.
The horizontal divergence of the bending magnet at BM05 is 2.4 mrad. We observed a 5.16 mm Si(333) diffraction spot at a distance of 7 m after the crystal from a 5 mm impinging beam. This means a divergence of the collimator of only 4.7 arcsec in the horizontal plane, which is much smaller compared with other crystal collimation methods (Ferrari & Korytar, 2001). The horizontal divergence is not coupled to the vertical divergence. The vertical divergence is set by the dispersive setting of the crystals, the of the DuMond diagram, and is independent of the divergence of the impinging beam.
4. Conclusions
Xray collimation as a novel application of diffractive–refractive optics was studied theoretically and experimentally. A setup with two crystals in a dispersive arrangement successfully demonstrated Xray collimation in the energy range 6.3–18.8 keV. The experimental results are close to the theoretical calculations of the spectral dependence of the focusing distance. To reach a better agreement between the theory of beam shaping through diffractive–refractive optics and our experiment, further theoretical and experimental research is necessary.
A minor drawback of the device is the higher harmonics contamination, which overlaps on the primary beam. We can overcome this problem by using a special crystal setup (Hrdý et al., 2011) or by placing a slit between the two crystals. The widths of the diffraction spots in Fig. 5 are experimental proof of the collimating optics. An interesting feature of this collimation method is its energy range. Being able to collimate lower energies (below 20 keV) can supplement compact refractive lenses. The degree of collimation is adjustable with a shifting of the value of the focusing distance at which we approximate collimation.
Acknowledgements
This work was supported by the following funds: Institutional Research Plan AS CR (No. AVOZ 10100522), MPO CR (FRTI1/412) and MSMT CR (INGO LA 10010 and MSM 0021622410). We would like to thank Mr Lukáš and the company Polovodiče a.s. which provided the crystals.
References
Artemiev, N., Hrdý, J., Peredkov, S. & Artemev, A. (2004). J. Synchrotron Rad. 11, 157–162. Web of Science CrossRef CAS IUCr Journals Google Scholar
Ferrari, C. & Korytar, D. (2001). J. Appl. Cryst. 34, 608–612. Web of Science CrossRef CAS IUCr Journals Google Scholar
Hrdý, J. (1998). J. Synchrotron Rad. 5, 1206–1210. Web of Science CrossRef IUCr Journals Google Scholar
Hrdý, J., Kuběna, A. & Mikulík, P. (2005). J. Phys. D, 38, 4325–4328. Web of Science CrossRef CAS Google Scholar
Hrdý, J., Mikulík, P. & Oberta, P. (2011). J. Synchrotron Rad. 18, 299–301. Web of Science CrossRef IUCr Journals Google Scholar
Hrdý, J., Mocella, V., Oberta, P., Peverini, L. & Potlovskiy, K. (2006). J. Synchrotron Rad. 13, 392–396. Web of Science CrossRef IUCr Journals Google Scholar
Hrdý, J. & Oberta, P. (2008). Rev. Sci. Instrum. 79, 073105. Web of Science PubMed Google Scholar
Hrdý, J. & Siddons, D. P. (1999). J. Synchrotron Rad. 6, 973–978. Web of Science CrossRef IUCr Journals Google Scholar
Mori, T. & Sasaki, S. (1995). Rev. Sci. Instrum. 66, 2171–2173. CrossRef CAS Web of Science Google Scholar
Oberta, P., Mikulík, P., Kittler, M., Hrdý, J. & Peverini, L. (2010). J. Synchrotron Rad. 17, 36–40. Web of Science CrossRef CAS IUCr Journals Google Scholar
Renninger, M. (1961). Z. Naturforsch. Teil A, 16, 1110–1111. Google Scholar
Snigirev, A., Kohn, V., Snigireva, I. & Lengeler, B. (1996). Nature (London), 384, 49–51. CrossRef CAS Web of Science Google Scholar
© International Union of Crystallography. Prior permission is not required to reproduce short quotations, tables and figures from this article, provided the original authors and source are cited. For more information, click here.